RigExpert Stick XPro Specifications
Frequency range: 0.1 – 1000 MHz
Frequency input step: 1 kHz
Measurements for 12.5, 25, 28, 37, 50, 75, 100, 150, 200, 300, 450 and 600Ω systems
SWR measurement range: from 1 to 100 in digital mode, from 1 to 10 in graph modes
Range R and X: 0… 2000, -2000… 2000
Operating modes: Multé (a quick check of multiband antennas), HAM (a quick check of single band antenna), Single (measure SWR, |Z|, R, X, C, L, RL, Mag&Phase), TDR (distance to fault), SWR chart, Return Loss chart, RX Chart, OSL calibration mode, Сable tools (StubTuner, Cable Length, Impedance, Loss measurements)
HF output: Connector type: N-type. Output signal shape: square, from 0.1 to 1000 MHz. Output power: -10 dBm (at a load of 50 Ohms).
Power: One Lithium-ion 18650 (included). Сharge time: 3 hours when using a charger. When the analyzer is connected to a PC or DC adapter with a USB connector, it receives power from these sources
Interface: TFT display, 220х220 pixels resolution. 2 × 3 keys on the waterproof keyboard. English (by default) and Japan menu and help screens. USB connection to a personal computer
Bluetooth: ver. 4.2 BLE Single-mode, Class B, via USB 2.0 Type-C
Dimensions: 185 x 40 x 33 mm (7.3 x 1.6 x 1.3 inches)
Operating temperature: -20°- 55°C (-4°-131°F)
Weight: 185 grams (6.5 Oz) with battery
Warranty: 2 years
RigExpert Stick XPro is made in Ukraine.
* All specifications are subject to change without notice or obligation.